Citation: Chauhan H, “Complete Solution for pMDI Products: From Development to Quality Control Testing”. ONdrugDelivery Magazine, Issue 85 (Apr 2018), pp 22-25.
Heli Chauhan outlines the major challenges encountered in testing pressurised metered dose inhalers, and how Proveris’ portfolio of instruments can be an invaluable addition to the testing workflow.
Pressurised metered dose inhalers (pMDIs) are currently the largest revenue-generating segment in the asthma and chronic obstructive pulmonary disease (COPD) drug delivery devices market and are expected to maintain this position until 2020.1 This results in extensive testing needs for pharmaceutical companies developing and manufacturing these products. The large volume of complex tests presents many challenges. With industry-leading instruments, support and lab services, Proveris provides a complete solution for pMDI testing throughout the life of the product.
KEY CHALLENGES IN PMDI TESTING
Automated actuation of devices eliminates the variability observed in manual actuation altogether. Vereo® actuators, controlled by Viota® software, ensure that the actuation parameters (velocity, stroke length, hold time) are controlled in a user-defined manner and are consistent across multiple actuations. Moreover, keeping the actuation parameters consistent throughout, across tests, results in high reproducibility of data. Proveris’ software platform offers complete traceability of all actuation events along with force-position-time profiles to aid in root-cause analysis during an out-of-specification (OOS) analysis.
The shake-to-fire delay time (i.e. time between shaking the device and its actuation) of a pMDI device is known to play a role in dose content variability.2 Furthermore, specific shaking profiles are required for certain products depending on the type and number of excipients present.3 This underlines the importance of determining and performing the proper shaking profile to ensure the correct dose is delivered every time.
“This lack of shaking could be fatal for some patients, should they get huge amount of drug in the initial shots from the device and mostly just propellant afterwards…”
The same effect as described above was observed when spray pattern (SP) area was measured using Proveris’ Sprayview® measurement system over the entire life of the device (Figure 3). Initial high SP area followed by a steep drop was observed for the “no shaking” condition. In comparison, a more uniform SP area was seen for the five-second shake duration.
These observations highlight the importance of shaking properly and reproducibly when testing pMDI devices both during development and as a quality control measure. Proveris instruments provide the user with the flexibility of programming the shaking angle, frequency, shaking duration and shake-to-fire delay so that all the devices have a uniform shaking profile for all the required tests. This will also eliminate shaking as a cause for inconsistencies in dose content and spray performance results.
PROVERIS AS A PMDI TESTING SOLUTION PROVIDER
Proveris aims to provide a complete solution for testing pMDI products with its precision instruments (Figure 4). The Vereo® SFMDx Automated Actuators are flexible and fit seamlessly into multiple testing workflows, such as:
- Shot weight measurement
- Dose content uniformity
- Aerodynamic particle size distribution (cascade impaction)
- Particle size distribution (laser diffraction)
- Spray pattern and plume geometry
- Fire-down of sprays between testing.
Spray Pattern and Plume Geometry as a Prescreening Tool for Clinical Trials
Spray pattern measurements from the SprayVIEW® instrument are a valuable screening tool during early development. The spray pattern is sensitive to changes in individual parameters, such as orifice length, orifice diameter and chamber depth, which are crucial design characteristics of the pMDI actuator.4 Gaining the ability to see the effects of these parameters on spray performance can have a significant impact on successful product development and prevent costly late-stage development failures.
Spray pattern and plume geometry measurements give a substantial amount of valuable information, irrespective of being a regulatory requirement by the US FDA. To illustrate this, take the measured spray pattern for two pMDI devices using the same canister and only a slight difference in the actuator. As displayed by Figure 5, the spray duration for the two are significantly different (60 ms versus 140 ms).
Further, the plume geometry data and the playback video of the plume from these devices gives a qualitative insight about the direction of the spray. Direction of spray is important to determine during development, as any skew can lead to higher deposition on the sides of the mouthpiece and low availability to the patient during clinical study, ultimately leading to failure in vivo. This quantitative and qualitative set of data can be invaluable for decision making prior to running an expensive clinical study. Proveris also offers lab services that include patient usage studies, device characterisation, formulation/device screening as well as qualitative/quantitative analysis of product performance. These studies can be especially helpful for companies who do not plan on testing in-house but nevertheless want to evaluate the product performance on a small scale or require consulting during early development.
Kinaero™ High-Throughput Fire-Down System
The newest addition to Proveris’ family of instruments is the Kinaero™ High-Throughput Fire-Down system for pMDIs. Firing down pMDI devices can be tedious and time consuming. Proveris’ Kinaero system addresses this issue, capable of firing down either canisters only or entire devices with actuators using specific easy to insert cassettes (Figure 6).
By replicating human use of inhalers, the Kinaero™ system provides reproducible actuation throughout product life. The software platform, with database storage and retrieval is compliant with 21CFR Part 11. Programmable shaking angle, frequency, duration and inter-actuation delay along with multiple modes of actuation (force, position or time based) provides greater flexibility during automated fire down. The compact benchtop model with a large touchscreen display offers the following key features and benefits:
- Automated fire down for up to 10 pMDI devices
- Multiple modes of actuation based on force, position and time
- Programmable shaking angle, frequency and duration
- Self-contained evacuation system that eliminates tedious waste disposal methods
- Updated operation and data management software with database storage and retrieval
- US 21 CFR Part 11 compliant.
“Automated actuation of devices eliminates the variability observed in manual actuation altogether. Vereo® actuators, controlled by Viota® software, ensure that the actuation parameters are controlled in a user-defined manner and are consistent across multiple actuations…”
Since the entire system is self-contained and requires only AC power, no vacuum or pressurised air connections are necessary. The multi-level filtration system containing high efficiency particulate air (HEPA) and activated carbon filters can withstand tens of thousands of shots before replacement is required. Moreover, no additional cleaning steps are required along with easy replacement of filters and system alerts to remind the user that a filter is due for replacement. The software platform also offers multiple operator levels with differences in privileges depending on whether the user is in an R&D or QC setting. Additional features such as the Break/Resume mode allows the user to pause a method, walk away from the instrument and resume it easily.
The Kinaero™ system fits into the pMDI testing workflow with ease, allowing users more control over their testing parameters. It also reduces the ergonomic burden on the analyst of firing down multiple devices. The time saved increases the operational efficiency of the lab and reduces testing backlog.
This article described some of the key challenges in pMDI testing, including the importance of consistent actuation and shaking parameters along with the need for traceability for OOS results. Proveris Scientific provides a solution to these issues with its portfolio of instruments that fit into most of the testing workflows, from shot-weight testing and spray characterisation to firing down of devices between tests. Consistent actuation and shaking profiles ensure accurate and reproducible actuation during each run. Along with supporting these workflows, spray pattern and plume geometry results from the SprayVIEW® instrument give insight into product performance early during development, thereby improving the chances for success during in vivo studies.
- “COPD and Asthma Devices Market is Expected to Reach $34.3 Billion, Global, by 2020”. Press Release, Allied Market Research, Nov 2014.
- Hautmann J et al, “Effect of Time between Actuation on the Dose Variability for Three Metered Dose Inhalers”. Proc RDD Europe 2013 (Berlin, Germany), 2013, Vol 2, pp 429–434.
- Newcomb A, Farina D, “Understanding the Importance and effects of shaking on pMDI performance”. Proveris Scientific Poster, DDL Conference (Edinburgh, UK) Dec 2015.
- Liao L et al, “Spray pattern: A rapid and sensitive early development tool for respiratory drug products”. Inhalation Magazine, Oct 2017.