Citation: Dierick W, Nakamura K, “Ready-To-Use Prefillable Syringes: Sterilisation Effects on Biopharmaceuticals”. ONdrugDelivery Magazine, Issue 71 (Oct 2016), pp 50-54.
William Dierick and Koji Nakamura report studies comparing the effects of different sterilisation on the extractables and leachables content of biotherapeutic products in prefilled syringes.
Pharmaceutical industry growth is buoyed this year by the expectation that 12 new drugs are forecast to reach blockbuster sales by 2020. Within this market, the biotech sector is a large contributor as indicated by the number of biopharmaceuticals within the top-10 drugs by worldwide sales in 2016. R&D in this space is forecast to continue at a high pace with four biotech products in the top-10 of the biggest launches in 2016.1
Therapeutic proteins are typically administered by injection and by using prefilled syringes (PFS). However, proteins may be sensitive to heat, oxidation and have the propensity to aggregate.2-4 Protein aggregation and the elicitation of anti-drug antibodies (ADAs) may have detrimental effects on drug efficacy, pharmacokinetics and safety. There is evidence that protein aggregation may enhance immunogenicity and this can be an important factor in causing adverse events.5-9 Immunogenicity has been reported with contributing factors related to excipients / formulation and interactions with contact materials of the primary drug container.10-12
Protein-aggregation factors derived from storage in PFS may include silicone oil, usually applied to improve the smooth gliding of the rubber plunger along the barrel, and tungsten oxide, a contaminant derived from the syringe glass manufacturing process.13-16
In 2014, these concerns led the US FDA to issue a Guidance for Industry: “Immunogenicity Assessment for Therapeutic Protein Products”. It states that “interactions between therapeutic protein products and the container closure may negatively affect product quality and immunogenicity. These interactions are more likely with prefilled syringes of therapeutic protein products”. In this Guidance, FDA recommends that sponsors should conduct a comprehensive extractables and leachables laboratory assessment using multiple analytical techniques to assess the attributes of the container-closure system that could interact with and degrade protein therapeutic products.17
With emphasis on the interrelated areas (drug product formulation – container closure system – manufacturing) for therapeutic proteins, the method of sterilisation for prefillable syringes may also become a key factor to mitigate interactions with therapeutic proteins. For sterile ready-to-fill syringes, typical methods of sterilisation include Ethylene Oxide (EtO) gas sterilisation, irradiation (gamma irradiation or e-beam) and steam sterilisation,
depending also on the applied materials used in the PFS system.
EXTRACTABLES / RESIDUES FROM DIFFERENT STERILISATION METHODS
New developments have been made in recent years to establish a low-leachable PFS system with a focus on application with therapeutic proteins; particularly PLAJEX™ a cyclo-olefin polymer (COP) PFS with i-coating™ plunger stoppers to eliminate the use of silicone oil as a lubricant of the syringe. These sterile ready-to-use syringes are steam sterilised within the tub/nest presentation.18-21
Terumo evaluated the material of PLAJEX™ prefillable syringes and compared the extractables (and residues) upon using different sterilisation methods. Sterilisation of the materials was conducted respectively with steam sterilisation (121°C for 30 min), EtO (EtO concentration 20%), and irradiation (25 kGy). Extractables were prepared by extractions with water for injection (WFI) at conditions of 121°C for 60 min. The levels of organic compounds in the extractable were determined by liquid chromatography-mass spectrometry (LC-MS) according to methods reported previously.20 Figure 1 shows the graphics of relative abundance obtained by LC-MS. Peaks were detected for irradiated samples whereas no peaks were detected for steam sterilisation and EtO sterilisation. In addition, residual EtO was monitored over time by gas chromatography using headspace sampling. Figure 2 shows the results of residual EtO over time of storage. The potential effect of residual EtO on biopharmaceuticals is not well known and will be subject to our additional research and experiments.
IRRADIATION: RADICAL GENERATION & INFLUENCE ON THERAPEUTIC PROTEINS
Sterilisation by irradiation is a technique commonly applied for polymer-based ready-to-fill syringes. Earlier publications indicated the occurrence of radicals from irradiated polymer-based PFS, affecting the level of protein oxidation as stored in these syringes.23-26
Additionally, a protein oxidation study was conducted where erythropoietin (EPO) was dissolved in an aqueous solution containing 2 mM Na2 HPO4 and 0.06 mg/mL polysorbate 80 to obtain a final concentration of EPO with 24,000 IU/mL. Analysis of oxidised methionine in the EPO solution was conducted in using HPLC. The results, shown in Figure 5, demonstrated that oxidation rate increased over time for the protein stored in prefillable syringes sterilised by irradiation. The oxidation rate of proteins stored in steam-sterilised prefillable syringes remains similar to that of non-sterilised syringes. Therefore, protein oxidation may be attributed to the generation of radicals present in the irradiated syringes.
It is generally believed that radicals persist for a very short period and more investigations were made therefore to verify the effects of time elapsing between irradiation and the time of filling of the tested protein solution (EPO solution). Samples were stored at 25°C prior to filling and filled syringes were also stored at 25°C for one month before oxidised methionine was quantified. Figure 6 shows the quantity of detected radicals over time and Figure 7 shows the protein oxidation ratio as measured for those time points. A decrease in oxidation ratio can be noted but the experiments suggest that the effect on the therapeutic proteins may persist over a longer period of storage time after sterilisation. Such effects can be mitigated by applying steam sterilisation for polymer ready-to-use syringes.
According to a report by Reigh et al, 27 radicals occur after polymer irradiation, and accelerate the continuous polymer oxidation. In this process, C=O or C-O bonds are produced (auto-oxidation). Development of this auto-oxidation cycle provides radicals to each molecule, resulting in a longer-term persistence of radicals. Our experiments reveal a similar reaction to the aforementioned auto-oxidation.
In consideration of the complexity and challenges associated with the development of therapeutic proteins, different regulatory guidance and directions have been developed over the years, emphasising the importance of assessing the possible interactions between biotherapeutics and the container closure system, as it may have detrimental effects on therapeutic protein quality and immunogencitiy.
PLAJEX™ with i-coating™ plunger stoppers creates a primary drug container that mitigates possible interactions from tungsten, silicone oil and aggregation thereof. Using steam sterilisation as the sterilisation method of ready-to-fill COP syringes may assist in mitigating interactions with the therapeutic protein by reducing the risks of protein oxidation.
- Urquhart L, Gardner J, Elmhirst E, “EP Vantage 2016 Preview”. Evaluate Ltd, December 2016.
- Jones NS, et al, “Silicone oil induced aggregation of proteins”. J Pharm. Sci, 2005, Vol 94(4), pp 918-927.
- Mahler HC, et al, “Protein aggregation: pathways, induction factors and analysis”. J Pharm Sci, 2009, Vol 98(9), pp 2909-2934.
- Wang W, et al, “Protein aggregation– pathways and influencing factors”. Int J Pharm, 2010, Vol 390(2), pp 89-99.
- Jenke D, et al, “Extractable/Leachable Substances from Plastic Materials Used as Pharmaceutical Product Containers/ Devices”. PDA J Pharm Sci Technol, 2002, Vol 56(6), pp 332-371.
- Jenke D, et al, “Linking Extractables and Leachables in Container/Closure Applications”. PDA J Pharm Sci Technol, 2005, Vol 59(4), pp 265-281.
- Markovic I, “Evaluation of safety and quality impact of extractable and leachable substances in therapeutic biologic protein products: a risk-based perspective”. Expert Opin Drug Saf, 2007, Vol 6(5), pp 487-491.
- Basant S, “Immunogenicity of therapeutic proteins. Part 2: Impact of container closures”. Biotechnol Adv, 2007, Vol 25(3), pp 318–324.
- Baker MP, et al, “Immunogenicity of protein therapeutics”. Self/Nonself, 2010, Vol 1(4), pp 314-322.
- Casadevall N, et al, “Pure redcell aplasia and antierythropoietin antibodies in patients treated with recombinant erythropoietin”. N Engl J Med, 2002, Vol 346(7), pp 469-475.
- Gershon SK, Luksenburg H, Cote TR, Braun MM, “Pure red-cell aplasia and recombinant erythropoietin”. N Engl J Med, 2002, Vol 346, pp 1584–1586.
- Locatelli F, et al, “Pure Red-Cell Aplasia “Epidemic” — Mystery Completely Revealed?” Perit Dial Int, 2006, Vol 27(Suppl2), pp S303-S307.
- Mensch CD, Davis HB, “Inhibition of Tungsten-Induced Protein Aggregation by Cetyl Trimethyl Ammonium Bromide”. PDA J Pharm Sci Technol, 2012, Vol 66, pp 2-11. (doi:10.5731/ pdajpst.2011.00806)
- Seidl A, et al, “Tungsten-Induced Denaturation and Aggregation of Epoetin Alfa During Primary Packaging as a Cause of Immunogenicity” Pharm Res, 2012, Vol 29(6), pp 1454-1467.
- Liu W, et al, “Root Cause Analysis of Tungsten-Induced Protein Aggregation in Pre-filled Syringes”. PDA J Pharm Sci Technol, 2010, Vol 64(1), pp 11-19.
- Jiang Y, “Tungsten-induced protein aggregation: Solution behavior”. J Pharm Sci, 2009, Vol 98(12), pp 4695-4710.
- US FDA, “Guidance for Industry – Immunogenicity Assessment for Therapeutic Protein Products”. August 2014.
- Yoshino K, et al, “Development of new prefillable syringe to solve pharmaceutical problems of biological drug formations”. Pharm Tech Japan, 2014, Vol 30(11), pp 47-52.
- Yoshino K, et al, “Functional evaluation and characterization of a newly developed silicone oil-free prefillable syringe system.” J Pharm Sci, 2014, Vol 103(5), pp 1520-1528.
- Kiminami H, et al, “Low Leachable Container System Consisting of a Polymer-Based Syringe with Chlorinated Isoprene Isobutene Rubber Plunger Stopper” PDA J Pharm Sci Technol, 2015, Vol 69(6), pp 713-724.
- Kiminami H, et al, “Development of the low-leachable prefillable syringe system”. Pharm Tech Japan, 2015, Vol 31(11), pp 37-40.
- Nakamura K, et al, “Assessment of the effects of sterilization methods on protein drug stability by elucidating decomposition mechanism and material analysis”. Int J Pharm, 2015, Vol 484(1-2), pp 51-56.
- Nakamura K, et al, “A strategy for the prevention of protein oxidation by drug product in polymer-based syringes”. PDA J Pharm Sci Technol, 2015, Vol 69(1), pp 88-95.
- Dierick W, et al, “A PolymerBased Prefillable Syringe System to Minimise Risk of Protein Oxidation”. ONdrugDelivery, October 2015, Issue 61, pp 52-56.
- Dierick W, et al, “Using Prefilled Syringes for Biopharmaceuticals: Development & Challenges”. ONdrugDelivery, February 2015, Issue 55, pp 10-16.
- “Methods of preparation of sterile products”. European Pharmacopoeia 8.0, Monograph 5.1.1.
- Reich L, Stivala S, et al, “Auto oxidation of hydrocarbons and polyolefins”. M. Dekker Inc, 1969
Previous articleTHE MANDREL CHALLENGE